The world's most trusted name in Durometers & Thermometers
0 item(s) in cart/ total: $0
A Scale Durometers
> PTC® Digital Durometer Shore A Scale 211A
Calibration Certificate Included with durometer
PTC® Digital Durometer Shore A Scale 211A
Add Digital Durometer SmartCable™
6 Foot Cable [+$345.00]
20 Foot Cable [+$395.00]
Hand Switch or Foot Switch for Smart Cable™
Hand Switch [+$75.00]
Foot Switch [+$75.00]
Put me on the Waiting List
Add to Cart
NIST/A2LA to ISO 17025 Calibration Certificate Included.
Need a Durometer Stand? See Compatible Stands Below.
Compatible Test Stands
The A Scale Durometers Measure the hardness of neoprene, EPDM rubber, polyurethane, printing rollers, silicone, white PVC and other similar materials.
The Pencil Style Digital Durometers with a 1/2" diameter base (1.3cm) and overall length will allow the durometer to be used in confined or hard to reach areas with 0.1 reading resolution. A knurled surface gives a more secure grip when making handheld readings.
All Digital Durometers come complete with Electronic Max Hold feature, Test Block, Carrying Case and a NIST traceable certification. PTC Metrology is accredited by A2LA to ISO/IEC 17025:2005 and ANSI Z-540.
NIST/A2LA to ISO 17025 Certification Included
1/2" Diameter Base - The Smallest Base Permitted by ASTM D2240
Max Hold Pointer Standard
LCD Display with 0.1 Point Resolution
Accessory Test Stands Available for all Models
Includes Precision Aluminum Test Block, Batteries, 115V AC Adaptor & Carrying Case
Meets or exceeds current ASTM D2240 Specs
Range (Type A) 0-100 points
Accuracy of ±1 Point
Height 6 in. (15.2 cm)
Width 2-1/4 in. (5.7 cm)
Depth 1-1/2 in. (3.8 cm)
Weight 13 oz. (170 g)
Shipping Weight 4lb (1.4kg)
For a complete calibration check of mainspring, and visual and mechanical check of indenter, the instrument should be returned to PTC® (see Guarantee & Calibration Service) or refer to ASTM D2240 Specifications. PTC® recommends the unit be returned at least every 12 months for this
check. For a quick field check, follow the guideline below. Under no circumstance should a test block be used as a standard to calibrate a durometer.
1. The pointer should read zero when no force is applied to the indenter of the durometer.
2. Hand hold the durometer and insert the indenter into the hole of the calibrated test block. Apply enough force to make firm contact between the top surface of the test block and the presser foot. The dial reading should agree with the value stamped on the check block (±1). Several tests should be made and the results averaged.
3. The indenter must protrude 0.098 to 0.100 inches below the presser foot.
4. When the indenter is fully displaced, the durometer should read 100 points. Use care as to not damage the tip of the indenter.
Hydraulic Test Stand 7000A
Deadweight Test Stand Model 476
Rubber Test Block Kit for ASTM Type A Durometers Model 400A
Digital Durometer USB SmartCable®
PTC® Durometer Stand for A | B | E | O Pencil Durometers Model 476
PTC® Hydraulic Durometer Stand for A | B | O Durometers Model 7000A
Recently Viewed Items
Browse Similar Items
Digital Pencil Durometer