The world's most trusted name
in Durometers & Thermometers
0 item(s) in cart/ total: $0
Shore A Scale Durometers
> PTC® e2000 Digital Durometer Shore A Scale 511A
NIST Traceable Digital Durometer Shore A Scale
PTC® e2000 Digital Durometer Shore A Scale 511A
Add Digital Durometer SmartCable™
6 Foot Cable [+$345.00]
20 Foot Cable [+$395.00]
Hand Switch or Foot Switch for Smart Cable™
Hand Switch [+$75.00]
Foot Switch [+$75.00]
Put me on the Waiting List
Add to Cart
Accredited by A2LA to ISO/IEC 17025, ANSI/NCSL Z540-1.
Made in the U.S.A.
Compatible Test Stands
A2LA ISO 17025 Calibration Certificate From PTC Metrology™ Included ($150.00 Value)
The Digital Durometer features a large direct read LCD display, electronic max hand, USB data ports for computer compatibility, a locking feature to prevent accidental erasure of tolerances, automatic shutoff, and low battery warning.
All PTC® digital durometers come complete with Test Block, Carrying Case and A2LA/NIST Traceable Calibration Certificate to ISO 17025 from
PTC®'s SmartCable can be added to any digital durometer. The output of the SmartCable USB Keyboard is a standard 2.0 keyboard signal. Readings can be sent to any Windows application's cursor point or highlighted cell. Works great with
. No special software required.
NIST/A2LA to ISO 17025 Certification Included
1.25" Diameter Base - The Optimal Size & Shape for Making Measurements
Max Hold Pointer Standard
LCD Display with 0.1 Point Resolution
Accessory Test Stands Available for all Models
Includes Precision Aluminum Test Block, Batteries, 115V AC Adaptor & Carrying Case
Meets or exceeds current ASTM D2240 Specs
Range (Type A) 0-100 points
Accuracy of ±1 Point
Height 6 in. (15.2 cm)
Width 2-1/4 in. (5.7 cm)
Depth 1-1/2 in. (3.8 cm)
Weight 13 oz. (170 g)
Shipping Weight 4lb (1.4kg)
For a complete calibration check of mainspring, and visual and mechanical check of indenter, the instrument should be returned to PTC® (see Guarantee & Calibration Service) or refer to ASTM D2240 Specifications. PTC® recommends the unit be returned at least every 12 months for this check. For a quick field check, follow the guideline below. Under no circumstance should a test block be used as a standard to calibrate a durometer.
1. The pointer should read zero when no force is applied to the indenter of the durometer.
2. Hand hold the durometer and insert the indenter into the hole of the calibrated test block. Apply enough force to make firm contact between the top surface of the test block and the presser foot. The dial reading should agree with the value stamped on the check block (±1). Several tests should be made and the results averaged.
3. The indenter must protrude 0.098 to 0.100 inches below the presser foot.
4. When the indenter is fully displaced, the durometer should read 100 points. Use care as to not damage the tip of the indenter.
Hydraulic Test Stand 7000A
Deadweight Test Stand 478
Rubber Test Block Kit for Durometers A, B and O Model 400A
Digital Durometer USB SmartCable®
Durometer Deadweight Test Stand (1 kg Weight) 478
PTC® Hydraulic Durometer Test Stand Model 7000A
PTC® Digital Durometer Shore A Scale 211A
Aluminum Test Block ASTM Type A
PTC® e2000 Durometer 'Shore A Scale' Model 501A
Recently Viewed Items
Browse Similar Items
Digital e2000 Durometer
Digital e2000 Durometer