The world's most trusted name in Durometers & Thermometers
0 item(s) in cart/ total: $0
C Scale Durometers
> PTC® Digital Pencil Durometer Shore C Scale Model 212C
212C Digital Pencil Durometer
PTC® Digital Pencil Durometer Shore C Scale Model 212C
Add Digital Durometer SmartCable™
6 Foot Cable [+$345.00]
20 Foot Cable [+$395.00]
Hand Switch or Foot Switch for Smart Cable™
Hand Switch [+$75.00]
Foot Switch [+$75.00]
Put me on the Waiting List
Add to Cart
NIST/A2LA to ISO 17025 Calibration Certificate Included
Made in the USA
Compatible Test Stands
C Scale Durometers Measure the Hardness of Medium Hard Rubber, Elastomers, and Thermoplastics.
The Pencil Style Digital Durometers with a 1/2" diameter base (1.3cm) and overall length will allow the durometer to be used in confined or hard to reach areas with 0.1 reading resolution.
All Digital Durometers come complete with Electronic Max Hold feature, Test Block, Carrying Case and a
NIST traceable certification
. PTC Metrology is accredited by A2LA to ISO/IEC 17025:2005 and ANSI Z-540.
NIST/A2LA to ISO 17025 Certification Included
1/2" Diameter Base - The Smallest Base Permitted by ASTM D2240
Max Hold Pointer Standard
LCD Display with 0.1 Point Resolution
Accessory Test Stands Available for all Models
Includes Precision Aluminum Test Block, Batteries, 115V AC Adaptor & Carrying Case
Meets or exceeds current ASTM D2240 Specs
Range (Type C) 0-100 points
Accuracy of ±1 Point
Height 6 in. (15.2 cm)
Width 2-1/4 in. (5.7 cm)
Depth 1-1/2 in. (3.8 cm)
Weight 13 oz. (170 g)
Shipping Weight 4lb (1.4kg)
For a complete calibration check of mainspring, and visual and mechanical check of indenter, the instrument should be returned to PTC® (see Guarantee & Calibration Service) or refer to ASTM D2240 Specifications. PTC® recommends the unit be returned at least every 12 months for this
check. For a quick field check, follow the guideline below. Under no circumstance should a test block be used as a standard to calibrate a durometer.
1. The pointer should read zero when no force is applied to the indenter of the durometer.
2. Hand hold the durometer and insert the indenter into the hole of the calibrated test block. Apply enough force to make firm contact between the top surface of the test block and the presser foot. The dial reading should agree with the value stamped on the check block (±1). Several tests
should be made and the results averaged.
3. The indenter must protrude 0.098 to 0.100 inches below the presser foot.
4. When the indenter is fully displaced, the durometer should read 100 points. Use care as to not damage the tip of the indenter.
Hydraulic Test Stand 7000D
Deadweight Test Stand 477
PTC® Hydraulic Durometer Stand for C | D | DO Durometers Model 7000D
PTC® Durometer Stand for C | D | DO Pencil Durometers Model 477
Aluminum Test Block ASTM Type C
Digital Durometer USB SmartCable®
PTC® e2000 Digital Durometer 'Shore C Scale' Model 512C
Recently Viewed Items
Browse Similar Items
Digital Pencil Durometer