The world's most trusted name
in Durometers & Thermometers
0 item(s) in cart/ total: $0
Shore D Scale Durometers
> PTC® Classic Durometer Gauge 'Shore D Scale' 307L
PTC® Shore D Scale Durometer Model 307L
PTC® Classic Durometer Gauge 'Shore D Scale' 307L
Do you require Certification to ISO/IEC 17025?
Add Durometer NIST/A2LA Certification [+$150.00]
No written certification required
Add Durometer Test Stand for Increased Repeatability
PTC® Model 475 Durometer Deadweight Test Stand [+$1,300.00]
PTC® Model 7000D Hydraulic Test Stand [+$2,800.00]
Put me on the Waiting List
Add to Cart
Compatible Test Stands
The Model 307L Type D Durometer is designed to measure ASTM (American Society for Testing and Materials) D2240 Type D hardness. These durometers can be used either with a test stand or hand-held.
Shore D Scale Durometers Test the Hardness of Pipeline Coating, Hard Rubber, Acrylic, Delrin, Kevlar, PVC, Nylon, Lexan, and Epoxies.
Specifications Sheet 307L
Max Hold Pointer Standard
Easy to Read 0 to 100 Point Dial
Accessory Test Stands Available
NIST/A2LA to ISO 17025 Certification Available
Includes Precision Aluminum Test Block
Meets or exceeds current ASTM D2240 Specs
Range (Type D) 0-100 points
Accuracy of ±1 Point
Height 4 in. (10.2 cm)
Width 2-1/4 in. (5.7 cm)
Depth 1-3/4 in. (4.4 cm)
Weight 8 oz. (227 g)
Shipping Weight 3lb (1.4kg)
For a complete calibration check of mainspring, and visual and mechanical check of indenter, the instrument should be returned to PTC® (see Guarantee &Calibration Service) or refer to ASTM D2240 Specifications. PTC® recommends the unit be returned at least every 12 months for this check. For a quick field check, follow the guideline below. Under no circumstance should a test block be used as a standard to calibrate a durometer.
1. The pointer should read zero when no force is applied to the indenter of the durometer.
2. Hand hold the durometer and insert the indenter into the hole of the calibrated test block. Apply enough force to make firm contact between the top surface of the test block and the base of the durometer. The dial reading should agree with the value stamped on the check block (±1). Several tests should be made and the results averaged.
3. The indenter must protrude 0.098 to 0.100 inches below the base of the durometer.
4. When the indenter is fully displaced, the durometer should read 100 points. Use care as to not damage the tip of the indenter.
320 Spring Load Cell Stand
475 5 kg Weight Stand
7000D Hydraulic Stand
Rubber Test Block Kit for Durometers D, C and DO Model 400D
Durometer Stand Spring Load Cell 320
Durometer Test Stand 5 kg Weight 475
PTC® Hydraulic Durometer Test Stand Model 7000D
Aluminum Test Block ASTM Type D
PTC® Shore D Ergo Durometer Gauge 409D
Recently Viewed Items
Browse Similar Items
Classic Style Durometers