The world's most trusted name
in Durometers & Thermometers
0 item(s) in cart/ total: $0
Shore D Scale Durometers
> PTC® Shore D Ergo Durometer Gauge 409D
PTC® Durometer (Shore D Scale) Model 409D
PTC® Shore D Ergo Durometer Gauge 409D
Do you require Certification to ISO/IEC 17025?
Add Durometer NIST/A2LA Certification [+$150.00]
No written certification required
Add Durometer Test Stand for increased repeatability
PTC® Model 475 Durometer Deadweight Test Stand [+$1,300.00]
PTC® Model 7000D Hydraulic Test Stand [+$2,800.00]
Put me on the Waiting List
Add to Cart
Accredited by A2LA to ISO/IEC 17025, ANSI/NCSL Z540-1.
Made in the U.S.A.
Compatible Test Stands
PTC®'s Model 409 ASTM Type D Durometer
measures the hardness of pipeline coating epoxies, plexiglass, epoxy, vinyl, cellulose acetate, and Kevlar. The favorite hand-held durometer of the pipeline coating inspectors.
Accurate readings can be obtained
in any position.
The instrument features a low glare serialized
dial with bold, easy to read numbers. The
durometer comes standard with a Max Hold hand
which maintains the peak reading until reset.
The case tapers to a base that is 1" x 1" to allow
better access to confined areas. A removable
top knob (8-32) threads permit both easy handheld
use or attachment to stands.
The Model 409 meets or exceeds the current ASTM D2240 Standards.
Specifications Sheet Durometer 409
Base Tapers to 1" x 1"
Max Hold Pointer Standard
Easy to Read 0 to 100 Point Dial
Accessory Test Stands Available
NIST/A2LA to ISO 17025 Certification Available
Includes Precision Aluminum Test Block
Meets or exceeds current ASTM D2240 Specs
Range (Type D) 0-100 points
Accuracy of ±1 Point
Height 4 in. (10.2 cm)
Width 2-1/4 in. (5.7 cm)
Depth 1-3/4 in. (4.4 cm)
Weight 6 oz. (170 g)
Shipping Weight 3lb (1.4kg)
For a complete calibration check of mainspring, and visual and mechanical check of indenter, the instrument should be returned to PTC® (see Guarantee & Calibration Service) or refer to ASTM D2240 Specifications. PTC® recommends the unit be returned at least every 12 months for this check. For a quick field check, follow the guideline below. Under no circumstance should a test block be used as a standard to calibrate a durometer.
1. The pointer should read zero when no force is applied to the indenter of the durometer.
2. Hand hold the durometer and insert the indenter into the hole of the calibrated test block. Apply enough force to make firm contact between the top surface of the test block and the base of the durometer. The dial reading should agree with the value stamped on the check block (±1). Several tests should be made and the results averaged.
3. The indenter must protrude 0.098 to 0.100 inches below the base of the durometer.
4. When the indenter is fully displaced, the durometer should read 100 points. Use care as to not damage the tip of the indenter.
320 Spring Load Cell Stand
475 5 kg Weight Stand
7000D Hydraulic Stand
Rubber Test Block Kit for Durometers D, C and DO Model 400D
Durometer Stand Spring Load Cell 320
Durometer Test Stand 5 kg Weight 475
PTC® Hydraulic Durometer Test Stand Model 7000D
PTC® Classic Durometer Gauge 'Shore D Scale' 307L
Aluminum Test Block ASTM Type D
PTC® e2000 Durometer 'Shore D Scale' Model 502D
PTC® Pencil Durometer Model 202D
PTC® e2000 Digital Durometer 'Shore D Scale' Model 512D
A2LA to ISO 17025 Certified
PTC® Digital Durometer Shore D Scale 212D
A2LA to ISO 17025 Certified
Recently Viewed Items
Browse Similar Items
Ergo Style Durometers