The world's most trusted name
in Durometers & Thermometers
0 item(s) in cart/ total: $0
Shore D Scale Durometers
> PTC® Ergo Durometer 'Shore D Scale' Model 409D
Shore D Scale Durometer Model 409D
PTC® Ergo Durometer 'Shore D Scale' Model 409D
Do you require Certification to ISO/IEC 17025?
Add Durometer NIST/A2LA Certification [+$150.00]
No written certification required
Put me on the Waiting List
Add to Cart
Email a friend
Accredited by A2LA to ISO/IEC 17025, ANSI/NCSL Z540-1.
Made in the U.S.A.
Compatible Test Stands
PTC®'s Model 409 ASTM Type D Durometer
measures the hardness of pipeline coating epoxies, plexiglass, epoxy, vinyl, cellulose acetate, and other similar materials. The favorite hand-held durometer of the pipeline coating inspectors.
Accurate readings can be obtained
in any position.
The instrument features a low glare serialized
dial with bold, easy to read numbers. The
durometer comes standard with a Max Hold hand
which maintains the peak reading until reset.
The case tapers to a base that is 1" x 1" to allow
better access to confined areas. A removable
top knob (8-32) threads permits both easy hand
held use or attachment to stands.
The Model 409 meets or exceeds American Society for Testing and Materials (ASTM) D2240 Specifications.
Base Tapers to 1" x 1"
Max Hold Pointer Standard
Easy to Read 0 to 100 Point Dial
Accessory Test Stands Available
NIST/A2LA to ISO 17025 Certification Available
Includes Precision Aluminum Test Block
Meets or exceeds current ASTM D2240 Specs
Range (Type D) 0-100 points
Accuracy of ±1 Point
Height 4 in. (10.2 cm)
Width 2-1/4 in. (5.7 cm)
Depth 1-3/4 in. (4.4 cm)
Weight 6 oz. (170 g)
Shipping Weight 3lb (1.4kg)
For a complete calibration check of mainspring, and visual and mechanical check of indenter, the instrument should be returned to PTC® (see Guarantee & Calibration Service) or refer to ASTM D2240 Specifications. PTC® recommends the unit be returned at least every 12 months for this check. For a quick field check, follow the guideline below. Under no circumstance should a test block be used as a standard to calibrate a durometer.
1. The pointer should read zero when no force is applied to the indenter of the durometer.
2. Hand hold the durometer and insert the indenter into the hole of the calibrated test block. Apply enough force to make firm contact between the top surface of the test block and the base of the durometer. The dial reading should agree with the value stamped on the check block (±1). Several tests should be made and the results averaged.
3. The indenter must protrude 0.098 to 0.100 inches below the base of the durometer.
4. When the indenter is fully displaced, the durometer should read 100 points. Use care as to not damage the tip of the indenter.
320 Spring Load Cell Stand
475 5 kg Weight Stand
7000D Hydraulic Stand
Rubber Test Block Kit for Durometers D, C and DO Model 400D
Durometer Spring Load Cell Stand 320
Durometer Test Stand 5 kg Weight 475
PTC® Hydraulic Durometer Test Stand Model 7000D
Aluminum Test Block ASTM Type D
PTC® e2000 Durometer 'Shore D Scale' Model 502D
PTC® Classic Durometer 'Shore D Scale' Model 307L
Recently Viewed Items
Browse Similar Items
Ergo Style Durometers