The world's most trusted name
in Durometers & Thermometers
0 item(s) in cart/ total: $0
Durometer Test Stands
Durometer Deadweight Test Stands
> Durometer Deadweight Test Stand 1 kg Weight Model 476
Model 476 Durometer Test Stand
Durometer Deadweight Test Stand 1 kg Weight Model 476
Put me on the Waiting List
Add to Cart
Accredited by A2LA to ISO/IEC 17025, ANSI/NCSL Z540-1.
Made in the U.S.A.
This stand is for use with
PTC®'s line of pencil style durometers
and is also fully compatible with durometers of other
The Precision engineered durometer stand has a calibrated 1kg deadweight to provide the ASTM D2240 recommended force for durometer types A, B, E and O. When properly used this stand and your durometer will make highly accurate hardness readings.
The support table is self-aligning for use with a non-parallel material.
ASTM D2240 Type 1 Specimen to Indenter
Test Stands Provide Correct Load for ASTM Types A, B & O
Increases Repeatability & Reduces User Error
Simplifies Hardness Testing
Holds PTC® Pencil Style Durometers and Other Common Pencil Style
Maximum height for sample 6 in. (15.2 cm).
Height: 17 in. (43.2 cm).
Width: 6 in. (15.2 cm).
Depth:6-1/2 in. (16.5 cm).
Shipping weight: 13 lbs. (11.3 kg).
Please read completely before operating your stand.
After carefully unpacking the unit, place the stand upright on desk or bench.
The model 476 is for Pencil Style Durometers.
Note: Always keep a small rubber or soft plastic block below the indentor to protect the tip from contacting the support table directly.
Using the rear knob adjust the height of the upper assembly near the top of the support rod. Please be careful as the assembly is heavy. Tighten the rear knob to secure the head assembly in place.
Use the included hex key to loosen the front mounting bracket thumb screw and swing open the front mounting bracket. Insert the Pencil Style Durometer so that the lower edge of the barrel adapter
is resting on the top surface of mounting bracket and close the front mounting bracket. Turn the thumb screw to secure durometer in place and then tighten with the hex key.
Place the appropriate weight on the 1/8 inch shaft at the top of the stand. For ASTM types A, B, E or O, use the 1 kg deadweight. For ASTM types C, D, or DO, use the 5 kg deadweight.
The durometer pressor foot can be set parallel to the support table or used in the self-aligning mode for measuring non-parallel material. Make 5 readings and average the test results.
When making a measurement, turn knob and lift stage so that sample contacts the durometer base and the durometer & deadweight assembly is lifted slightly. See durometer instructions for additional information on testing and reporting test results.
The durometer pressor foot can be set parallel to the support table or used in the self-aligning mode for non-parallel material. Make 5 readings and average the test results.
PTC® Pencil Durometer 'Shore O Scale' Model 201/O
PTC® Pencil Durometer Model 201A
PTC® Pencil Durometer 'Shore B Scale' Model 201B
PTC® Digital Durometer Shore A Scale 211A
PTC® Digital Durometer Shore B Scale 211B
PTC® Digital Pencil Durometer Shore O Scale Model 211/O
Recently Viewed Items